CO (Carbon Monoxide Mixing Ratio System) Handbook [electronic resource].
- Washington, D.C. : United States. Dept. of Energy. Office of Environmental Management, 2011.
Oak Ridge, Tenn. : Distributed by the Office of Scientific and Technical Information, U.S. Dept. of Energy.
- Additional Creators:
- Atmospheric Radiation Measurement Program (U.S.), United States. Department of Energy. Office of Environmental Management, and United States. Department of Energy. Office of Scientific and Technical Information
- Restrictions on Access:
- Free-to-read Unrestricted online access
- The main function of the CO instrument is to provide continuous accurate measurements of carbon monoxide mixing ratio at the ARM SGP Central Facility (CF) 60-meter tower (36.607 °N, 97.489 °W, 314 meters above sea level). The essential feature of the control and data acquisition system is to record signals from a Thermo Electron 48C and periodically calibrate out zero and span drifts in the instrument using the combination of a CO scrubber and two concentrations of span gas (100 and 300 ppb CO in air). The system was deployed on May 25, 2005.
- Report Numbers:
- E 1.99:doe/sc-arm/tr-072
- Published through SciTech Connect.
- Funding Information:
View MARC record | catkey: 14128988