Qualification of the Lasentec M600P Particle Size Analyzer and the Red Valve Model 1151 Pressure Sensor [electronic resource].
- Washington, D.C. : United States. Dept. of Energy, 2000. and Oak Ridge, Tenn. : Distributed by the Office of Scientific and Technical Information, U.S. Dept. of Energy.
- Physical Description:
- 58 pages : digital, PDF file
- Additional Creators:
- Pacific Northwest National Laboratory (U.S.), United States. Department of Energy, and United States. Department of Energy. Office of Scientific and Technical Information
- Restrictions on Access:
- Free-to-read Unrestricted online access
- The Lasentec M600 in-line particle size analyzer was installed at Oak Ridge National Laboratory (ORNL) in August 1998 to support retrieval of the Gunite and Associated Tanks (GAAT). Before installation at ORNL, the sensor underwent validation testing at the Pacific Northwest National Laboratory (PNNL) Instrument Validation facility. Mechanically, the instrument worked well during validation testing and met all expectations. Operationally, much was learned about optimum ways to display and interpret the data. Slurry samples taken during the in-line tests at PNNL were shipped to the vendor for analysis with a benchtop Lasentec sensor. These experiments were performed to determine if off-line analyses yield particle size distributions similar to those generated by the in-line sensor. It was determined that the Lasentec sensor measures repeatable chord lengths as long as particles are ''presenter'' to the sensor window the same way. After the initial non-radioactive simulant testing at PNNL, the instrument was shipped for radioactive validation and acceptance testing in the Slurry Monitoring Test System (SMTS) connected to the Tank W-9 of the GAATs at ORNL. For all acceptance tests conducted at ORNL, the variation in the chord length distribution and the total particle count corresponded very well with the slurry density data as determined using an in-line Promass 63M Coriolis meter. Based on the performance results obtained, the Lasentec M600P FBRM is expected to meet the requirements for measuring the particle size distribution during the slurry transfer operations at Hanford and the Oak Ridge GAAT remediation project. The Red Valve pressure sensor was endorsed at the Hanford Site following instrument validation tests at PNNL and is currently in operation in the Tank 241-C-106 pump pit. While this instrument measures pressure within a transfer line, this type of pressure sensor could be configured to measure pressure drop over time. In turn, the status of a slurry transfer could be inferred from the pressure-drop measurement. In 1998, four Red Valve pressure sensors (with Sensotech Model AE-213 pressure transducers) were installed before and after the booster pumps of the 4-in. slurry (SL-200) and supernatant (SN-200) transfer lines between Tank 241-C-106 and Tank 241-AY-1 02. These pressure sensors have been in operation for over 1 year, and to date, the sensors have been trouble-flee according to the operators involved with slurry and supernatant transfer operations. Based on these observations, it is apparent that the Red Valve pressure sensors could be installed at the end of the slurry transfer lines and used to measure the pressure drop in the system.
- Published through SciTech Connect., 01/28/2000., "pnnl-13064", "820201000", and NG Colton; JR Bontha; EA Daymo; TD Hylton; CK Bayne; TH May.
- Funding Information:
- AC06-76RL01830 and 29042
View MARC record | catkey: 14412411