Intial synchroscan streak camera imaging at the A0 photoinjector [electronic resource].
- Washington, D.C. : United States. Dept. of Energy, 2008.
Oak Ridge, Tenn. : Distributed by the Office of Scientific and Technical Information, U.S. Dept. of Energy.
- Physical Description:
- 5 pages : digital, PDF file
- Additional Creators:
- Fermi National Accelerator Laboratory, United States. Department of Energy, and United States. Department of Energy. Office of Scientific and Technical Information
- Restrictions on Access:
- Free-to-read Unrestricted online access
- At the Fermilab A0 photoinjector facility, bunch-length measurements of the laser micropulse and the e-beam micropulse have been done in the past with a single-sweep module of the Hamamatsu C5680 streak camera with an intrinsic shot-to-shot trigger jitter of 10 to 20 ps. We have upgraded the camera system with the synchroscan module tuned to 81.25 MHz to provide synchronous summing capability with less than 1.5-ps FWHM trigger jitter and a phase-locked delay box to provide phase stability of ≈1 ps over 10s of minutes. This allowed us to measure both the UV laser pulse train at 244 nm and the e-beam via optical transition radiation (OTR). Due to the low electron beam energies and OTR signals, we typically summed over 50 micropulses with 1 nC per micropulse. We also did electron beam bunch length vs. micropulse charge measurements to identify a significant e-beam micropulse elongation from 10 to 30 ps (FWHM) for charges from 1 to 4.6 nC. This effect is attributed to space-charge effects in the PC gun as reproduced by ASTRA calculations. Chromatic temporal dispersion effects in the optics were also characterized and will be reported.
- Report Numbers:
- E 1.99:fermilab-conf-08-107-ad
- Other Subject(s):
- Published through SciTech Connect.
Presented at 13th Beam Instrumentation Workshop (BIW08), Lake Tahoe, California, 4-8 May 2008.
Ruan, J.; Lumpkin, A.H.
- Funding Information:
View MARC record | catkey: 14680369