Sonde Adjust Value-Added Product Technical Report [electronic resource].
- Washington, D.C. : United States. Dept. of Energy. Office of Science, 2012. and Oak Ridge, Tenn. : Distributed by the Office of Scientific and Technical Information, U.S. Dept. of Energy.
- Additional Creators:
- Atmospheric Radiation Measurement Program (U.S.), United States. Department of Energy. Office of Science, and United States. Department of Energy. Office of Scientific and Technical Information
- Restrictions on Access:
- Free-to-read Unrestricted online access
- The Sonde Adjust (SONDEADJUST) value-added product (VAP) creates a file that includes all fields from original Atmospheric Radiation Measurement Climate Research Facility (ARM Facility) radiosonde files and contains several value-added fields that provide adjustments related to well-known humidity issues. SONDEADJUST produces data that correct documented biases in radiosonde humidity measurements. Previous efforts towards applying some of these corrections are available via the discontinued PI product sgpsondecorr1miloC1. Unique fields contained within this datastream include smoothed original relative humidity, dry bias corrected relative humidity, and final corrected relative humidity. The smoothed RH field refines the relative humidity from integers-the resolution of the instrument-to fractions of a percent. This profile is then used to calculate the dry bias corrected field. The final correction fixes the time-lag problem and uses the dry-bias field as input into the algorithm. In addition to dry bias, solar heating is another correction that is encompassed in the final corrected RH field. Output from SONDEADJUST differs from the previous RH-corrected datastreams in important ways. First, all three types of ARM radiosondes-Vaisala RS-80, RS-90, and RS-92-are corrected using dedicated procedures and/or parameters. Second, the output variables include all of those found in the original radiosonde file: dry bulb temperature, dewpoint temperature, wind speed, wind direction, eastward wind component, northward wind component, wind status (a Vaisala-produced field used in conjunction with the Loran system), ascent rate, and original relative humidity. Additional humidity fields are smoothed relative humidity, dry biased corrected relative humidity, final ambient relative humidity, and scaled adjusted relative humidity. Third, quality control (QC) flags of the fields from the original radiosonde datastream are brought into the SONDEADJUST output file. Additional QC variables are created for the new fields.
- Published through SciTech Connect., 01/09/2012., "doe/sc-arm-tr-102", and Troyan, D.
- Funding Information:
View MARC record | catkey: 14795330